- Monday - Sunday: 09:00 - 22:00
95% Particle Penetration Filtering Efficiency, Filter class provides >99.9% Viral and Bacterial Filtering Efficiency, Safe and Comfortable Breathing Resistance, Valves Tested for inward valve leakage.
VOGNXT VM CONFORMS TO THESE STANDARDS IN TESTING:
The filter efficiency as stated in 42 CFR Part 84.181 is a minimum efficiency for each filter of ≥ 95% ( ≤ 1% penetration) . The test articles submitted by the sponsor conform to the criteria for filter efficiency.
NOTE: Vognxt is marketed for general use directly to consumers and is NOT CERTIFIED by NIOSH/CDC/NPPTL for workplace where an OSHA Compliant respiratory protection program to include Fit Testing on each individual is required.
Vognxt is manufactured under Quality Plan at ISO Certified plant in Seoul, Korea. Masks manufactured in India are approved for sales by India Ministry Safety Certification.
Summary of testing and Certifications:
VOGNXT BRAND GUIDELINES: TESTING, TECHNICAL
Important Note: Vognxt is designed as filtering face mask for helping to protect from particles. Our filtering masks are designed to present to the public an important and efficient tool for protecting health.
Vognxt CONFORMS in testing to standards established for workplace masks.
The filtering efficiency of a mask depends on a proper fit. This is why we offer Vognxt in multiple sizes. Also Vognxt is Quantitatively Fit Tested on human subjects to make sure the filtering efficiency is retained over a series of motions like walking, head movements, bending over, talking, etc.
Vognxt measures well within the safety range for breathing resistance and is therefore comfortable to wear and safe for wearing for long stretches of time.
QUALITY MANAGEMENT SYSTEM: Established and maintains a Quality Management System (QMS), which includes a Quality Manual and detailed standard operating procedures. Vognxt operates under the QMS to ensure its business processes are focused on consistently meeting customer requirements and enhancing customer satisfaction of the Vognxt brand. Vognxt strives to maintain a high level of product quality and, through its manufacturing partners, seeks to continuously improve its products.
PRODUCT CONFORMS to standards in testing at independent testing labs.
95% Particle Penetration Filtering Efficiency
The inhalation resistance criteria as stated in 42 CFR Part 84.180 is an initial inhalation not exceeding 35 mm water column height pressure. The test articles submitted by the sponsor conform to this criterion for airflow resistance. The exhalation resistance criteria as stated in 42 CFR Part 84.180 is an initial exhalation not exceeding 25 mm water column height pressure.
Determination of Valve Leakage
According to 42 CFR Part 84.182 the leakage between the valve and valve seat shall not exceed 30 mL/min. The test articles submitted by the sponsor conform to the criteria for exhalation valve leakage.
Viral Filtration Efficiency
>99.9% There were no detected plaques on any of the Andersen sampler plates for this test article.
Bacterial Filtering Efficiency
>99.9% There were no detected colonies on any of the Andersen sampler plates for this test article.